Analysis of Haloacetic Acids in Tap Water Using LC/MS/MS [LCMS-8030]Aplikace | 2012 | ShimadzuInstrumentationLC/MS, LC/MS/MS, LC/QQQIndustriesŽivotní prostředíManufacturerShimadzuKey wordshaloacetic, mcaa, tcaa, dcaa, acids, drinking, acid, dryng, blank, water, min, recovery, news, area, monochloroacetic, dichloroacetic, trichloroacetic, concentration, method, requested, establishment, flow, concerning, tap, extraction, suggested, solvent, proposed, without, accordance, nebulizing, currently, eliminates, conducted, derivatization, probe, interference, temperature, transition, contaminants, steps, limitation, warranties, mass, phases, produced, kind, warranty, according, addition
Determination of Haloacetic Acids in Drinking Water by LC/MS/MS
Application NoteEnvironmentalDetermination of Haloacetic Acids inDrinking Water by LC/MS/MSAuthorsClaudimir Lucio do LagoDepartment of FundamentalChemistry,Institute of Chemistry,University of São Paulo, BrazilDaniela DanielAgilent Technologies, Inc.AbstractA fast, simple, and sensitive direct injection LC/MS/MS method has been developedfor the determination of nine haloacetic acids...
counts, countshaloacetic, haloaceticbdcaa, bdcaadrinking, drinkingmcaa, mcaatcaa, tcaawater, watercdbaa, cdbaambaa, mbaatbaa, tbaadbaa, dbaadcaa, dcaabcaa, bcaalcmrl, lcmrlchlorate
Analysis of Haloacetic Acids in Drinking Water Using Triple Quadrupole LC/MS/MS (LCMS-8050)
LAAN-A-LM-E066ApplicationNewsLiquid Chromatography Mass SpectrometryAnalysis of Haloacetic Acids in Drinking WaterUsing Triple Quadrupole LC/MS/MS (LCMS-8050)C95No.Haloacetic acids (HAAs), by-products of waterdisinfection, are formed from naturally occurringorganic and inorganic materials in water which reactwith the disinfectants chlorine and chloramine. TheJapanese Ministry of Health,...
recovery, recoverymcaa, mcaatcaa, tcaadcaa, dcaawater, waterdrinking, drinkinghaas, haasaverage, averagehaloacetic, haloacetictap, tapblank, blankmrm, mrmrate, ratespike, spiketerion
High-Speed Analysis of 9 Haloacetic Acids in Tap Water Using Triple Quadrupole LC/MS/MS
ApplicationNewsLiquid Chromatograph Mass Spectrometer LCMS™-8060High-Speed Analysis of 9 Haloacetic Acids in TapWater Using Triple Quadrupole LC/MS/MSNo. C234H. HoriikeUser Benefits Analysis is possible in only 15 minutes thanks to simple sample preparation requiring only dechlorination. In addition to 3 items listed...
origin, originpass, passdoes, doestap, tapmin, minbcaa, bcaambaa, mbaatbaa, tbaadbaa, dbaamcaa, mcaatcaa, tcaarecovery, recoveryacid, aciddcaa, dcaadbcaa
Analysis of 9 Haloacetic Acids in Tap Water Using the LCMS-8040 Triple Quadrupole Mass Spectrometer
LAAN-A-LM-E060ApplicationNewsLC/MSLiquid Chromatography Mass SpectrometryAnalysis of 9 Haloacetic Acids in Tap Water Using theLCMS-8040 Triple Quadrupole Mass SpectrometerC89No.Haloacetic acids are produced as byproducts ofchlorination during water treatment purificationprocesses. Three haloacetic acids are subject to tapwater quality standards with established limits,monochloroacetic acid...
bcaa, bcaambaa, mbaatbaa, tbaadbaa, dbaamcaa, mcaatcaa, tcaadcaa, dcaaspiked, spikedblank, blanksample, samplecdbaa, cdbaabdcaa, bdcaatap, tapacid, acidmin