A Sensitive Microflow LC-MS/MS Method for the Analysis of Fluticasone Propionate in Human Plasma
Applications | 2017 | WatersInstrumentation
LC/MS, LC/MS/MS, LC/QQQIndustries
fluticasone, propionate, microflow, human, plasma, ionkey, sensitive, system, robustness, note, application, enviroments, method, waters, dirtiness, xevo, analysis, configured, scale, extraordinarily, assay, ikey, sensitivity, uplc, acquity, initial, configuration, blocked, class, pharmacokinetics, curve, chromatography, quantitation, peak, implementing, over, targetlynx, solutions, bioanalytical, comprised, area, correctly, define, masslynx, ranging, conc, calibration, serial, injecting, facilitates
PEPTIDE AND PROTEIN BIOANALYSIS
[ APPLICATION NOTEBOOK ]PEPTIDE AND PROTEINBIOANALYSIS[ OUR SCIENTISTS ]Yun Alelyunas, PhDBefore coming to Waters in 2012, YunAlelyunas was a principal scientist and teamleader at AstraZeneca for 20 years whereshe was involved in industrial chemicalanalysis, in vitro ADME screening, andproject core...
ionkey, ionkeypeptide, peptideplasma, plasmaquantification, quantificationhuman, humanarea, areaxevo, xevoinsulin, insulinprotein, proteinuplc, uplcpeptides, peptidesmrm, mrmproteinworks, proteinworksantibody, antibodyusing
Increasing Throughput for the Analysis of Human Insulin and Related Biotherapeutic Analogs using the ionKey/MS System
[ TECHNOLOGY BRIEF ]Increasing Throughput for the Analysis of Human Insulin andRelated Biotherapeutic Analogs using the ionKey/MS SystemMichael Donegan, James Murphy, and Erin ChambersWaters Corporation, Milford, MA, USAThe ionKey/MS System enables the flexibility tomodulate between ultra-sensitivity and higherthroughput with ease.GOALTo...
insulin, insulinmicroflow, microflowionkey, ionkeyanalogs, analogsglulisine, glulisinehuman, humanbrief, briefdetemir, detemiraspart, aspartsystem, systemlispro, lisproglargine, glarginemodulate, modulatethroughput, throughputsensitivity
ionkey/MS - APPLICATION COMPENDIUM
[ ion key / MS ]APPLICATION COMPENDIUMDear Colleague,The 2014 introduction of the ionKey/MS System was a turning point for LC-MS.The promise of increased levels of sensitivity from smaller sample sizes was finally a reality. Wewere confident we had designed a...
ionkey, ionkeyikey, ikeyuplc, uplcsystem, systemplasma, plasmasensitivity, sensitivityhuman, humanion, ionusing, usingmobility, mobilityarea, areadevice, deviceassay, assayseparation, separationquantification
Improving the Detection of Thyroglobulin in Human Plasma for Clinical Research by Combining SISCAPA Enrichment and Microflow LC-MS
Improving the Detection of Thyroglobulin in Human Plasma for ClinicalResearch by Combining SISCAPA Enrichment and Microflow LC-MSJay S. Johnson,1 Morteza Razavi,2 James Murphy,1 Selena Larkin,2 and Paul Rainville11Waters Corporation, Milford, MA, USA; 2 SISCAPA Assay Technologies, Washington, DC, USAA P...
fsp, fspmicroflow, microflowsiscapa, siscapahyroglobulin, hyroglobulinionkey, ionkeyikey, ikeyplasma, plasmapump, pumphuman, humanpeptide, peptideclinical, clinicaltrapping, trappingresearch, researchimproving, improvingtrap